Report all problems whether fixed or not to 3-LIVE (3-5483).
Error Code 1523 Could not pull the history off the unit!!!
You get a invalid line error if the line is not inserted in the right hole for the line. It doesn’t like that.
Error Code 1208
2/1/13: Unit was powered up and showed 3/4 battery power. Unit opened auto syringe correctly. No error code was given. Researched error Code 1208. It is Motor drive recognition of direction of Rotation. In other words the stepper motor was moved manually (Auto syringe pulled out or pushed in manually). Stepper motor is reset when unit is turned off and back on. Auto syringe returns to 0 point and resets. Unit was cleaned and returned to service LP
Error Code 2202
1/31/2014: Unit was powered up and showed full charge. Unit showed error 2202 which is stepmotor PH1A not 0. That refers to the motor that opens the door for the infusion tubing. The door was open but the unit believed the motor was closed. Left door open and restarted unit. Pressed door open button. Unit opened the locking mechanism and gave error again. Restarted unit again and unit cleared error. The stepper motor was reset when unit is turned off and back on. Unit was tested on 1200 ml/hr and infused 80 ml. Performed test again. Unit performed correctly both tests, was cleaned and returned to service. LP
5/30/2013: Unit was powered up and showed full charge. Unit powered up correctly. Tested unit by infusing 40 ml at 400ml/min. Then infused 60 ml at 200ml/hr. Both tests were performed on battery power to insure batteries were working correctly. Could not duplicate problem of unit not infusing. Unit performed both tests correctly, was cleaned and returned to service. LP
Bianca was doing a functional test of two of the Perfusor (Syringe) pumps that came back from repair. She noticed that when a 60 ml B-D syringe that one of the possible selections was a Neomed 60 ml syringe. It was not recognized from testing the pumps before so everything was reloaded onto these pumps and retested. With the syringe data reloaded the Neomed syringe was no longer available. We will have to be careful when we get a pump back to be sure that our settings are all correct. BBraun do put a notice on the pump when the drug library is lost but they do not mention any possible changes in the disposable syringe data.
Return from B-Braun
1/31/2014: Unit returned from repair at B-Braun. Unit was charged for > 24 hours. Unit was powered up and showed Full battery power. Re-installed Drug Library, Mod data and Disposable data on unit using the HiBase software. Tests were performed on battery power to verify battery condition. Unit opened auto syringe. No error code was given. Inserted syringe and infused 60mL over 10 minutes. Unit performed correctly, was cleaned as needed and returned to service. LP
1/31/2014: Performed physical, visual inspection on unit. Verified unit was performing correctly. No EST was performed as unit is isolated. Abbreviated PM was performed at this time(1 Year) as manufacturer suggests PM needs to be performed every 2 years. Device passed all tests, PM sticker updated and unit was entered into service. LP
The syringe data is in the disposable data so we should reload everything to be safe. On the infusomat we would load the library and the modification data and on the perfusor we would load the library, modification data, and disposable data. The process for doing this can be found on the K drive. The document is named “How to Load the Configuration and Library to BBraun Pumps”.
5/30/13 Reprogrammed unit with correct drug data, disposable, and modification data. Unit was cleaned as needed and returned to service. LP
Drug Library (DL 0104130)
When doing Multiples—Make sure plugged in with black cable (Not White)
Be sure to plug into P2 connector.
Read from Device You can select all Data for Multiple Pumps.
Write to device
File- Find File -- it will be a .DLE file
Send to device
Mod Data—Menu Settings Etc
Date Time= Remove the pumps that you have programmed from the White box.
Date & Time
Green button- Set
Turn pump on set to English US
Double Click on pump
Worker ID = any 4 digits
Calibration data OK
Check lock bolt
Hit blue button on pump
Is lock bolt open = Yes
Is lock bolt cleared a& symbol flashing = Yes
Tab on Right == Disconnect
Invalid Code Error = Data Lock
When doing Multiple Pumps, data is stored in the Depot Folder
C:\B.Braun Space\HiBaSeD_5.1.0\Spreadsheat folder. It is saved under its serial Number.
K:\Bio-MED\BBraun Hist\CN ##### SN #####\
C:\B.Braun space\HiBaSeD_5.1.0\Depot\Infusomat Space S ***Serial number****
Checking Alarm—History Do the Print, then Save as .PDF and save to ?? folder.
Do NOT press save. Data is saved when you pull it in